Once I bought a popular “Chinese” volt- and ammeter gauge with 100V/10A ratings. Unfortunately, I didn’t find any good resource about how to wire/connect it. Actually, it’s not rocket science, but I had waited for it 2-3 weeks and didn’t want to blow it at first try. All the diagrams were somehow too technical for me.
I updated this article in 2018 August.
The original article included only one variant of these digital volt- and ammeters. Now I have included three most common variants of them. If you order digital 100V/10A volt- and ammeter gauge from China or other markets – it’s most likely one of following.
They all have a bit different wire colors. I will cover wiring for all of them in this post.
100V / 10A DC volt- and ammeters with differently colored wires
These meters are good when you want to measure current and voltage at the same time. A feature that usual multimeters lack. For example, I have used it to measure the output of my DIY solar panel. They are inexpensive and easy to connect.
How do these ammeters work?
These meters calculate current flow by the voltage drop across the shunt resistor. They have built-in shunts that should withstand measuring as much as 10 amperes. You must see shunt behind your volt- and ammeter. That small bridge bent from a thick wire on PCB.
Built-in shunt on volt- and ammeter-100V / 10A / DSN-VC288
Important! If there is none- this guide is not for you. You need a wiring diagram with an external shunt instead. Also when measuring current that exceeds 10A which can be handled with the internal one.
They are basically the same. There is a connector with thick wires and connector with thin wires. But coloring of wires vary. Amperage measurement is done through passing power through thick wires. Power to the meter itself is fed through thin wires.
They all can be wired such way that you don’t need a separate power supply for the electronics part. You can use the same power supply where your measured load gets electricity. But then you can only use a maximum of 30V since it’s maximum operating voltage for the meter itself. Also, you can’t measure voltages below 4.5V. If using the same power supply you even can leave thin black wire un-connected since they have common ground/GND.
Red and black thick wires + thin yellow, black and red
This one makes the most sense to me and it has model number DSN-VC288 marked on the back.
10A / 100V volt- and ammeter wiring with thick red, black wires and thin black, red, and yellow
Example on Amazon
Red, black and blue thick wires + thin black and red
This exact one was in my original article. Not very popular anymore I guess since can’t find many offers for these.
Example on Amazon
Red, black and yellow thick wires + red and black thin wires
Important! I have not tested this one myself. If someone can confirm this is correct I would be very thankful. From the info I found from sellers and one youtube video- this one has a bit tricky coloring. Voltage sense line is also a thick wire and it’s not yellow. It’s a red one instead!
Can someone please confirm this schematic is correct?
10A / 100V volt- and ammeter wiring with thick red, black and yellow wires and thin black and red
Example on Amazon
Here are some photos of testing them
Wired and testing with load
Wired and testing with load- overview
One test with the external power supply
Volt ammeter with external power supply
10A / 100V volt- and ammeter testing, DSN-VC288